MINOLTA 조도계 T-10A/T-10MA
조도 및 적산조도 측정용 단순 조도계
최대 30 point까지 측정 헤드 확장 가능 - Multi 조도측정시스템 구현
Illuminance Meter T-10A/T-10MA
JIS AA Class 및 DIN Class B 규격에 부합하는 조도계
쉽고 저렴한 Multi-Point 측정 구현(2~30 point)
방수헤드 적용 제품 공급 가능 - T-10WSA, T-10WLA(Waterproof mini receptor head)
Illuminance meter class
Relative spectral response
User calibration function
Illuminance Meter T-10A(Standard receptor), T-10MA(Mini receptor head)
Multi-function digital illuminance meter with detachable receptor head (Multi-point measurements or 2 to 30 points is possible)
Conforms to requirements for Class AA of JIS C
1609-1: 2006 "Illuminance Meter Part 1: General measuring instruments"
Conforms to DIN 5032 Part 7 Class B
Within 6%(f1') of the CIE spectral luminous efficiency V(λ)
T-10A, T-10MA: Within 3%, T-10WSA, T-10WLA: Within 10%
Auto range(5 manual ranges at the time of analog output)
Illuminance(lx), illuminance difference(lx), illuminance ratio(%), integrated illuminance (lx·h), integration time(h), average illuminance(lx)
T-10A, T-10MA: 0.01 to 299,900 lx; 0.001 to 29,990 fcd
T-10WSA, T-10WLA: 1.00 to 299,900 lx; 0.1 to 29,990 fcd*2
0.01 to 999,900 x 103lx·h 0.001 to 99,990 x 103 fcd·h/0.001 to 9999h
CCF (Color Correction Factor) setting function: Measurement value x 0.500 to 2.000
±2%±1 digit of displayed value
1mV/digit, 3V at maximum reading; Output impedance: 10㏀; 90% response time: 28ms
3 or 4 Significant-digit LCD with backlight illumination (Automatic illumination)
AC adapter AC-A308 (optional; for 1 to 10 receptors) or
AC adapter AC-A311 (optional; for 1 to 30 receptors)
72 hours or longer (when alkaline batteries are used) in continuous measurement
T-10A, T-10MA: -10 to 40℃, relative humidity 85% or less(at 35℃) with no condensation
T-10WSA, T-10WLA: 5 to 40℃, relative humidity of 85% or less(at 35℃) with no condensation
T-10A, T-10MA: -20 to 55℃, relative humidity 85% or less(at 35℃) with no condensation
T-10WSA, T-10WLA: 0 to 55℃, relative humidity of 85% or less(at 35℃) with no condensation
69 x 174 x 35mm
* 1 Conforms to requirements for Class AA ofJIS C 1609-1: 2006 for all items except cosine response(f2).
* 2 Although measurements below 1.00lx are possible, they may not be stable due to the effects of electricalnoise.
* Do not touch the cable during measurements.Doing so may result in unstable measurement values.
* Secure the cable during measurements.Failure to do so may result in unstable measurement values.